Resonant scattering for open port use
UHV chamber for soft x-ray scattering experiments with in-vacuum CCD detector.
For more details and current status of the experimental station contact the station scientist(s).List of publications
|Temperature range||low-temperature sample holder 6.5 - 300 K, high temperature sample holder 80 - 650 K|
|Pressure Range||5 x 10-8 mbar (regular), < 2 x 10-9 mbar (after bakeout)|
|Detector||In-vacuum CCD, photo diode, APD|
|Sample holder compatibility||Own system, can accommodate samples of various shapes. Please contact us beforehand.|
MethodsXMLD, XMCD, Time-resolved studies, NEXAFS, EXAFS, Reflectometry, Surface Diffraction, Single Crystal Diffraction, WAXS, REXS, Magnetic Scattering
- Different detectors including in vacuum CCD, Si diode, APD detectors can be rotated around the sample.
- Different sample stages for low temperatures (6.5 K ... 300 K) without sample transfer, and either 20 K ... 400 K or 80 K ... 650 K with sample transfer system.
- Motorized x,y,z,th,chi (+- 3 deg) manipulator, manual phi rotation.
- Motorized vertical translation of all detectors.
- Control software SPEC.